The account Probe Replacement settings define the time interval when probes are to be replaced. OneVue intelligently tracks the lifespan each OneVue Probe with CertiTrak by calculating the days remaining to be replaced based on the Probe Replacement settings. The days remaining to replace are provided in the Probe Replacement report. It's recommended these settings meet your organization’s policies.
Probe Replacement settings
The account settings set the Default value for all probes. If the settings vary for a specific probe, they can also be set from the sensor profile.
Probe Replacement: the time interval when a probe is to be replaced.
The default value is Every 2 Years; options include Every Year, Every 2 Years, Every 5 Years, and Every 10 years.
Replace Based On: the start date of the probe lifespan.
The First Use Date is the date the probe was first inserted into a sensor device. The Last Testing Date is the date of calibration documented in its Calibration Data Report. Both of these dates are stored in OneVue and displayed in the profile of sensor within its probe life section.
Go to Admin > Edit "your account name".
From the Probe Replacement section, configure the settings.
Probe Replacement: default value is Every 2 Years; options include Every Year, Every 2 Years, Every 5 Years, and Every 10 years.
Replace Based On: default value is First Use Date; options include First Use Date and Last Testing Date.
To apply a setting change, select Save.