Sensor assigned to a Monitored Asset is reporting a low battery level.
Temperature, Temperature and Humidity, and Differential Pressure Sensors: From the start date of this state, estimated battery life remaining is 30 days. This estimation is based on the use of Lithium 1.5V batteries, 1 hour Logging Interval, 8 hour Check-In Interval, and adequate wireless signal.
Water Leak and Contact Closure Sensors: From the start date of this state, estimated backup battery life remaining is 7 days. This estimation is based on the use of Lithium 1.5V batteries, 1 hour Logging Interval, 8 hour Check-In Interval, and adequate wireless signal.
All or some of the below symptoms may be present.
Monitored Asset (assigned sensor) is in an Alarm state with a Low Battery status.
Low Battery alert notification received.
Sensor device yellow LED indicator is flashing and LCD displaying Low Battery icon.
To stop alert notifications, acknowledge the alert.
Replace the batteries. Insert the 3.0v Primex Lithium/Iron Disulfide Battery Pack or two 1.5v Lithium AA batteries. Follow the symbols showing the correct way to position the positive (+) and negative (-) ends of the battery pack.
Open the device cover and verify the battery switch is set to on (up position).
If also operating on AC power, verify the power adapter is plugged in. The green LED is illuminated when powered by AC.
Press and release the check-in button to clear the Low Battery status and Alarm state.