Primex sensors are equipped with a reading alarm audio alert. The audio alert is activated when a reading is out of range and the time period it has been out of range has exceeded the time period set in its Alarm Delay. Be default, the audio alert is disabled.
The sensor emits a continuous beep at one second intervals during a reading Alarm state. In addition to the audio alert, the device's red LED flashes once per second during a reading Alarm state.
The audio alert may be silenced by pressing and quickly releasing the silence button (bell icon) located on the front the sensor. When the silence button is pressed during an audio alert, the alert is NOT acknowledged in OneVue. When the audio alert resumes is dependent on the sensor's Resume Audio Alert After setting.
If the audio alert is disabled, the red LED flashes once per second during a reading Alarm state. Pressing the silence button (bell icon) suspends the red LED from flashing for the duration of the Alarm state.
Go to Devices > Sensors.
Select the Gateway ID link of the sensor.
Select Show Advanced Options.
To enable the audio alert, select the Enable Audio Alert checkbox. A checkmark indicates the audio alert is enabled. To disable the audio alert, deselect the Enable Audio Alert checkbox.
From the Resume Audio Alert After drop-down menu, select the duration of time the audio alert should resume after the silence button is pressed during a reading Alarm state.
The Update Pending status is set to Yes when a setting change is scheduled to be downloaded to the device during its next check-in to OneVue. Setting changes take effect when downloaded to the device. Once downloaded, its Updating Pending status is set to No.